867041 - CO2 Extension Intellivue Module
Model Number: 867041
The Philips 867041 module comes standard with CO2 Microstream. It is capable of invasive blood pressure, CCO, and temperature monitoring with purchased option.
The Philips 867041 CO2 extension module cannot be used in a stand alone capacity. It will not directly connect to a Philips monitor. It must be connected to a MMS, X2, MP2, MMX, X3 to work properly. This module has MSL connectors going to the monitor and then an additional connector to the patient module. Sidestream CO2 comes standard and you can add invasive pressure, temp, and CCO if needed. We see sidestream CO2 failures including occlusion, leak test failure, and CO2 equipment malfunctions. We also see communication or power issues.
- No communication with Monitor
- Software not compadible
- CO2 equipment malfunction
- CO2 Occlusion
- CO2 extention Equipment Malfunction
- Parameter malfunction
- Parameter blanked out
- Parameter connector damaged